ULT Technical Support
Ultrasound Leakage Tester, Technical Support
ULT Technical Information
- ULT User Manual
- ULT Training Guide
- ULT Software User Guide
- How to perform an Ultrasound Leakage Test
- ULT Technical Applicatiion Note
- ULT Adapter Cross Reference Guide
How to purchase ULT Testers and Accessories
Specialty ULT Testers
ULT Common Errors and Troubleshooting
|Error||Cause / Solution|
|Source Voltage too low||Internal voltage supply error. |
-If on battery power, try a new LITHIUM 9V battery.
NOTE: Alkaline Batteries Will Not Work Correctly.
-Try running the test with battery-only, and power supplyonly to narrow the problem.
|Internal Leakage too low|| |
Internal voltage supply error.
|Internal Leakage too high||Possible damaged Conductivity Probe|
|Conductivity too low||Conductive Bath error |
-Conductivity Probe not connected and/or submerged.
-Soak Tray or Soak Tube is not filled.
-Cleaning / Test solution is not conductive.
|Leakage too low||-Transducer not submerged enough/correctly. |
-Incorrect limit settings for that transducer.
-Cleaning solution drained during the test.
|Leakage too high||-Incorrect limit settings for that transducer |
How to Perform an Ultrasound Leakage Test
How to customize your ULT-2010 and ULT-2020
How to perform an ultrasound leakage test on GE Probes
How to perform an Ultrasound Leakage Test on SonoSite Probes
The ULT can be operated under battery power utilizing a 9V Lithium battery, or AC line operated utilizing the proper ULT Battery Eliminator.
|9V Lithium Battery||ULT Battery Eliminator|
- Do Not use Alkaline batteries.
- The ULT does Not charge the 9V Lithium battery.